Bioharness™ multivariable monitoring device. Part I: Validity.
20210281 ST [electronic version only]
Johnstone, J.A. Ford, P.A. Hughes, G. Watson, T. & Garrett, A.T.
Journal of Sports Science and Medicine, Vol. 11 (2012), No. 3 (September), p. 400-408, ref.
|Samenvatting||The Bioharness™ monitoring system may provide physiological information on human performance but there is limited information on its validity. The objective of this study was to assess the validity of all 5 Bioharness™ variables using a laboratory based treadmill protocol. 22 healthy males participated. Heart rate (HR), Breathing Frequency (BF) and Accelerometry (ACC) precision were assessed during a discontinuous incremental (0-12 km·h-1) treadmill protocol. Infra-red skin temperature (ST) was assessed during a 45 min-1 sub-maximal cycle ergometer test, completed twice, with environmental temperature controlled at 20 ± 0.1 °C and 30 ± 0.1 °C. Posture (P) was assessed using a tilt table moved through 160°. Adopted precision of measurement devices were; HR: Polar T31 (Polar Electro), BF: Spirometer (Cortex Metalyser), ACC: Oxygen expenditure (Cortex Metalyser), ST: Skin thermistors (Grant Instruments), P:Goniometer (Leighton Flexometer). Strong relationships (r = .89 to .99, p < 0.01) were reported for HR, BF, ACC and P. Limits of agreement identified differences in HR (-3.05 ± 32.20 b·min-1), BF (-3.46 ± 43.70 br·min-1) and P (0.20 ± 2.62°). ST established a moderate relationships (-0.61 ± 1.98 °C; r = 0.76, p < 0.01). Higher velocities on the treadmill decreased the precision of measurement, especially HR and BF. Global results suggest that the BioharressTM is a valid multivariable monitoring device within the laboratory environment. (Author/publisher)|
|Full-text||Beschikbaar Niet beschikbaar, klik om contact op te nemen voor een digitalisatie verzoek|
|Suggestie?||Neem contact op met de SWOV bibliotheek voor uw opmerkingen|